What are optical survey prisms?
Optical survey prisms are glass reflectors used with total stations and AMTS monitoring systems. Prisms reflect the infrared beam emitted by the total station, allowing it to calculate the precise distance to the prism and resolve the horizontal and vertical angles to the prism.
How are survey prisms used?
In monitoring applications, survey prisms are installed at monitoring points and control points.
Monitoring points are specified locations on objects that are subject to displacement, such as soldier piles, buildings, retaining walls, road surfaces, and railroad tracks. Survey prisms at monitoring points provide settlement and displacement data.
Control points are stable locations outside the zone of influence. Survey prisms installed at control points are used to measure the position of the AMTS itself at every observation cycle.
Installing and monitoring survey prisms
AMTS observation cycles are slowed by searching and retries. Optimize performance with careful placement of prisms and appropriate use of search windows and modes.
- Seek locations for prisms that keep sight lines free from possible interference, such as wire fences, building corners, reflective signage, and foliage.
- Install prisms in a way that provides as much angular separation as possible because lines of sight tend to converge with distance.
- Set search windows no larger than necessary. The AMTS begins searches with its detection beam at full power and may incorrectly lock onto reflections from traffic cones, road signs, and other prisms if the window is too wide.
- Set the appropriate search mode. When possible, use a mode that searches for expected shapes of the prism (circles or 360). If shapes cannot be resolved, use a mode that seeks hotspots in the return signal.
Types of Survey Prisms
The L-Bar Prism is installed on tunnel walls, building facades, foundation slabs, retaining walls, and almost any other location. It is used for both monitoring points and control points.
The Rail Clip is installed directly on the base of a rail and holds an L-Bar prism. The clip is available in 5.5 and 6 inch versions. Other versions can be installed in the ballast or on rail ties.